Home > Textbooks > Lessons In Electric Circuits > Vol. II - AC > AC Metering Circuits > Power Measurement

Chapter 12: AC METERING CIRCUITS

# Power Measurement

Power measurement in AC circuits can be quite a bit more complex than
with DC circuits for the simple reason that phase shift complicates the
matter beyond multiplying voltage by current figures obtained with
meters. What is needed is an instrument able to determine the product
(multiplication) of *instantaneous* voltage and current.
Fortunately, the common electrodynamometer movement with its stationary
and moving coil does a fine job of this.

Three phase power measurement can be accomplished using two dynamometer movements with a common shaft linking the two moving coils together so that a single pointer registers power on a meter movement scale. This, obviously, makes for a rather expensive and complex movement mechanism, but it is a workable solution.

An ingenious method of deriving an electronic power meter (one that generates an electric signal representing power in the system rather than merely move a pointer) is based on the Hall effect. The Hall effect is an unusual effect first noticed by E. H. Hall in 1879, whereby a voltage is generated along the width of a current-carrying conductor exposed to a perpendicular magnetic field: (Figure below)

*Hall effect: Voltage is proportional to current and strength of the perpendicular magnetic field.*

The voltage generated across the width of the flat, rectangular conductor is directly proportional to both the magnitude of the current through it and the strength of the magnetic field. Mathematically, it is a product (multiplication) of these two variables. The amount of “Hall Voltage” produced for any given set of conditions also depends on the type of material used for the flat, rectangular conductor. It has been found that specially prepared “semiconductor” materials produce a greater Hall voltage than do metals, and so modern Hall Effect devices are made of these.

It makes sense then that if we were to build a device using a
Hall-effect sensor where the current through the conductor was pushed by
AC voltage from an external circuit and the magnetic field was set up
by a pair or wire coils energized by the current of the AC power
circuit, the Hall voltage would be in direct proportion to the multiple
of circuit current and voltage. Having no mass to move (unlike an
electromechanical movement), this device is able to provide *instantaneous* power measurement: (Figure below)

*Hall effect power sensor measures instantaneous power.*

Not only will the output voltage of the Hall effect device be the
representation of instantaneous power at any point in time, but it will
also be a DC signal! This is because the Hall voltage polarity is
dependent upon *both* the polarity of the magnetic field and the
direction of current through the conductor. If both current direction
and magnetic field polarity reverses -- as it would ever half-cycle of
the AC power -- the output voltage polarity will stay the same.

If voltage and current in the power circuit are 90^{o} out of phase (a power factor of zero, meaning *no*
real power delivered to the load), the alternate peaks of Hall device
current and magnetic field will never coincide with each other: when one
is at its peak, the other will be zero. At those points in time, the
Hall output voltage will likewise be zero, being the product
(multiplication) of current and magnetic field strength. Between those
points in time, the Hall output voltage will fluctuate equally between
positive and negative, generating a signal corresponding to the
instantaneous absorption and release of power through the reactive load.
The net DC output voltage will be zero, indicating zero true power in
the circuit.

Any phase shift between voltage and current in the power circuit less than 90^{o}
will result in a Hall output voltage that oscillates between positive
and negative, but spends more time positive than negative. Consequently
there will be a net DC output voltage. Conditioned through a low-pass
filter circuit, this net DC voltage can be separated from the AC mixed
with it, the final output signal registered on a sensitive DC meter
movement.

Often it is useful to have a meter to totalize power usage over a period
of time rather than instantaneously. The output of such a meter can be
set in units of Joules, or total energy consumed, since *power* is a measure of work being done *per* unit time. Or, more commonly, the output of the meter can be set in units of Watt-Hours.

Mechanical means for measuring Watt-Hours are usually centered around the concept of the motor: build an AC motor that spins at a rate of speed proportional to the instantaneous power in a circuit, then have that motor turn an “odometer” style counting mechanism to keep a running total of energy consumed. The “motor” used in these meters has a rotor made of a thin aluminum disk, with the rotating magnetic field established by sets of coils energized by line voltage and load current so that the rotational speed of the disk is dependent on both voltage and current.